## PicoScope 5000 Series

#### FlexRes® Oscilloscopes

High speed and high resolution. Breakthrough ADC technology switches from 8 to 16 bits in the same oscilloscope.

# Measuring angular velocity

## Introduction

A small model airplane gets most of its forward thrust by a propeller. The propeller is driven by either an internal combustion engine or electric motor. The angular velocity, or speed at which the propeller revolves, is generally measured in Revolutions Per Minute — RPM — with the thrust of the airplane being directly proportional to the RPM of the propeller.

Usually the RPM of a model airplane is measured using a tachometer. This experiment shows how the RPM can be measured using a DrDAQ data logger.

### Equipment required

• DrDAQ data logger
• PicoScope oscilloscope software
• Model airplane
• Small flashlight / torch

## Experiment setup

For this experiment a small model airplane with two propellers, each with two blades, was used. The idea is to use the light sensor of the DrDAQ to measure the frequency of a revolving propeller. A simple flashlight is used as a light source. As the propeller consists of two blades the beam of light from the flashlight will be interrupted two times with each revolution of the propeller.

DrDAQ is used to measure the frequency (f) in hertz (cycles per second) of the light beam, and from this the RPM can be calculated:

RPM of propeller = (frequency/number of propeller blades) * number of seconds in a minute = (f/2) * 60

The meter function in PicoScope can display the frequency in hertz, or it can be calculated manually by measuring the displayed waveform. In this case the frequency would be:

frequency = (1/T) * 1,000,000 — where T is the time taken in µs by one complete cycle of a waveform.

To measure the RPM the model airplane was tied to a chair in an upright position so that the propeller is faced downward (as shown in Figure 1). The flashlight was fixed to the top, while the DrDAQ data logger was kept at the bottom. The distance of DrDAQ was adjusted by placing it on several books, with care taken not to place it too close to the propeller.

Figure 1: Measuring the RPM of a propeller using DrDAQ

In the pictures above the flashlight is shown very near to the propeller only for demonstration purposes. In actual practice, the flashlight should be kept higher and adjusted so that a clean shadow of the propeller is formed. To prevent any possibly interference that could case measurement errors, all nearby artificial lights such as tube lights, lamps, etc should be switched off.

When the equipment was correctly positioned, the light source is switched on and the vertical scale in PicoScope is adjusted so as to show a clear line. Then, using the remote control of the aeroplane, the propeller is started. After a short time the displayed waveform should stabilize. At this stage the waveform and the meter can be stopped and the results noted.

## Further study

This experiment could be extended to fully investigate how the diameter of the propeller, together with the number of blades, blade pitch and RPM affect the aircraft’s thrust.

Alternatively, a variation of this experiment could be to use a small DC electric motor and invesitgate how the RPM of the motor is related to the current flowing the armature of the motor.

## Testimonials

• The kit (2408B) is of obvious quality, easy to setup and calibrate and the free to download software has a reasonable learning curve. Superb kit, superb support, what more can I say.

Rop Honnor
• I have been using my 4224 PicoScope for years. I travel abroad so this has been ideal due to its physical size. Storage of waveforms on my Laptop is very easy allowing me to quickly email waveforms to my Colleagues.

Andrew
• Not many USB scopes works on Win & Mac & Linux too, so that proves me that guys from Pico really cares about us, customers. This made my decision much easier when I was looking to buy an USB scope.

Raul Trifan
• We have been using Picoscope 6404D for quite some time, and are amazed by its accuracy and powerful emulations while working with numerous signal evaluations.

J Mohanty
• PicoLog TC-08: This is a very nice unit that works consistently and reliably.

Jeff Hulett
• It is a great scope. I had a weird problem - it did not work on one of my PC’s. Customer service gave me first class service. If I could give 6 stars for customer service - I would do so.

Niels Larsen
• Perfect Partner for Development of Encoder controlled Stepper Motor Actuators. Since the included software is really stable, this type of device is a great tool for great tasks!

Helmut Schoettner
• A superb piece of equipment worth its weight in gold

Nigel Clinch
• So simple to use & beats any other I have ever used hands down.

John D Samsing
• Rather than webcam our benchtop scope screen, it was suggested we use a PicoScope to share waveforms via screen share.

We were a bit hesitant at first … but it worked flawlessly, and everybody is now out shopping for PicoScopes!

T Lövskog • SVP of Innovation at Additude AB
• Great functionality in a compact size. I really like moving the mouse pointer to a position and having the Time and Voltage display the values at that point. Calibration equipment is a breeze with that feature.

Don Horein

PC oscilloscope and data logger products