Post any questions you may have about our current range of oscilloscopes
I am using a PS3205 since April 05 and now the CHA is fixed at 20V. When I use the offset null tool, the screen shows a fixed line at zero.
Nevertheless, channel B is working OK.
Is there anything I can do or is this a hardware fault?
- Active User
- Posts: 17
- Joined: Thu Nov 11, 2004 2:40 pm
Thank you for your post
It appears your scope has developed a fault.
You can return it to us for repair. If you would like to do this, please contact us direct at: firstname.lastname@example.org
Michael - Tech Support
Web Support Forum
- Zen Master
- Posts: 656
- Joined: Thu Jul 07, 2005 12:41 pm
- Location: St Neots, Cambridgeshire
Return to USB PC Oscilloscopes
Who is online
Users browsing this forum: No registered users and 0 guests