Post any questions you may have about our current range of oscilloscopes
Can I set an amplitude of an AWG between -250 mV +250 mV in a pico-scope 2203?
I probe that in PicoScope6 with a pico-scope 2203 I can't. Is this a software fail? Can I set the amplitude of an AWG lower than 250 mV using the API?
If i use an arbitrary AWG I can set amplitude near to 0, isn't it?
- Posts: 0
- Joined: Fri Dec 16, 2011 2:17 pm
You can't set a value between -250mV and +250mV, this is a hardware limit and applies for Picoscope6 as well as software written to use the dll.
- Site Admin
- Posts: 2287
- Joined: Fri Jun 10, 2011 8:15 am
- Location: St. Neots
Return to USB PC Oscilloscopes
Who is online
Users browsing this forum: No registered users and 0 guests