I am using the PicoVNA 106 for measurements together with other measurement systems, and it seems like the VNA keeps on sending signals even after its single sweep measurement. I noticed this because the other measurement systems struggle to connect when the VNA is active, meaning there are some interfering signals coming from it. However, when the VNA is turned off (from the power supply), everything works as planned.
I am using the python library "picosdk-picovna-python-examples" and tried to find a function, which turns off or mutes the TX signal, but I could not find such thing. Is there a possibility to completely mute the TX Port?
Thanks in advance and best regards,
Discussions forum for our Sampling Oscilloscopes, VNA, and Pulse Generators
1 post • Page 1 of 1