I kind of busted the A-channel input on my picoscope 3205,
my inductive pickup (for ignition testing) was picking up a bit too much.
I don't have schematics for the unit unfortunately, but perhaps someone here know some common fault of an overload?http://rfelektronik.se/temp/div/Dsc_1006.jpg
Would it be the reed relays? Or the SST5912 gain block, or could it be the HA4600CBZ buffer?
Or the A/D's? But it looks like the other circuits would take the biggest hit...