Post any questions you may have about our current range of oscilloscopes
we have put a too hight tension on the input (picoscope 3205).
It breaks the IC SST5912 (Dual N channel JFET).
Could you tell me where we can buy it, or give us a reference of an equivalent product.
- Posts: 0
- Joined: Wed Nov 24, 2010 8:02 am
The SST5912 dual FET is difficult to source independently and even if you find one the channel offset may have to be adjusted afterwards. If would be best to send your scope back to us for repair. Please contact us on Help-desk:-email@example.com
- Zen Master
- Posts: 435
- Joined: Thu Jun 07, 2007 10:53 am
- Location: Cambridgeshire
Return to USB PC Oscilloscopes
Who is online
Users browsing this forum: No registered users and 0 guests