OK, angry mode.
I've just decided to give my DrDaq another go, as, after the fiasco i had last year with the Ph probe i gave up and bought a 10 quid ph probe.
Well, its packed in so i decided to get the Drdaq out as my fish are important to me.
the (replacement) ph probe was/is still in its little bottle and still wet. So i set it up. Well, the reading are STILL way off the mark.
Ph4 calibration solution reads at 7.29 apx and Ph7 calibration solution read about 6.93.
If i short the pins out, Ph7.
This is a NEW probe. Folks, twice ive had this problem and its not good for a 130 quid bit of kit to fail like this, especially when a 10 quid prob reads the calibration solutions at 3.96 and 7.04 respectively.
Please dont tell me the probe has dried out, it hasnt and has never been out of its bottle.
I recall a post where another user had had ther same issue and had altered some driver from an earlier version which solved the issue.
Does anyone have any ideas??? Else, i have a useless DrDaq for sale.