Hi,

I would like to estimate the error of temp. measurements.

I am using a K-thermocouple with a +/- 2.2 °C error

The TC-08 datasheet says:

Temperature accuracy: Sum of ±0.2 % of reading and ±0.5 °C

1) is the percentage calculated from the absolute temperature (in K) or from the temperature difference between the thermocouple measurement and the cold junction (ambient temp)?

2) does this value include the thermocouple error? (I doubt it)

I thought to calculate the total error as follows:

for measuring T=300°C in a room at 25°C.

dT=sqrt( (0.2*(300-25)/100 + 0.5)^2 + 2.2^2 ) = +/- 2.4°C

is that correct?

Thank you,

Raul

## TC-08 Accuracy

### Re: TC-08 Accuracy

Hi,

Yes, the cold junction (ambient) temperature is taken as the starting point for calculation of scaling errors, just as you have done in your equation. Thermocouple error is in addition to the TC-08 accuracy and from what I can remember your equation to calculate total error is correct.

Regards,

PeterF.

Yes, the cold junction (ambient) temperature is taken as the starting point for calculation of scaling errors, just as you have done in your equation. Thermocouple error is in addition to the TC-08 accuracy and from what I can remember your equation to calculate total error is correct.

Regards,

PeterF.