Thank-you for your inquiry with regard to your 2200 series application.
You can set an action for the program whenever there is an "Alarm" condition. One of the actions can be "Save Current Buffer" or "Save All Buffers" to the Hard Drive. We don't yet have an alarm caused by a trigger so you need to artificially cause an alarm when the signal triggers. Using the "Masks" function, it is easy to set up a mask which will always trigger an alarm on your particular waveform. In this way, you can set-up an equivalent to the "save on Trigger" command. This will have to do until we do a formal implementation of this function. Please contact again if you need more help getting it to work.
Mask limit testing is a feature that tells you when a waveform or spectrum goes outside a specified area, called a mask, drawn on the scope view or spectrum view. PicoScope can draw the mask automatically by tracing a captured waveform, or you can draw it manually. Mask limit testing is useful for spotting intermittent errors during debugging, and for finding faulty units during production testing.
To begin, go to the main PicoScope menu and select Tools > Masks > Add Masks. This opens the Mask Library dialog. When you have selected, loaded or created a mask.
Shows the allowed area (in white) and the disallowed area (in blue). Right-clicking the mask area and selecting the Edit Mask command takes you to the Edit Mask dialog. You can change the mask colors with the Tools > Preferences > Colors dialog; add, remove and save masks using the Masks menu; and hide and display masks using the Views > Masks menu.
(B) Failed waveforms
If the waveform enters the disallowed area, it is counted as a failure. The part of the waveform that caused the failure is highlighted, and persists on the display until the capture is restarted.
(C) Measurements table
The number of failures since the start of the current scope run is shown in the Measurements table. You can clear the failure count by stopping and restarting the capture using the Start/Stop button. The measurements table can display other measurements at the same time as the mask failure count.
To download the latest version of PicoScope 6 please proceed to the following linkhttp://www.picotech.com/software.html
Incorporating features such as: -
Merged channel options button and channel label
Automatic device reconnection
Buffer overview navigate on mask fails
Added bitstream to AWG creator
If you require further assistance please do not hesitate to contact via firstname.lastname@example.org