Post any questions you may have about our current range of USB data loggers
I have the (PP266) current clamp connected to the ADC 20. I have cut the plug off and connected the cable to channel 8 and AG.
However, i cannot get the clamp to measure any current.
Can anyone help?
Are you trying to measure AC current? The ADC-20 is not fast enough to capture 50Hz. If you are measuring DC current then you should be getting 1mV read on the ADC-20 for each Amp through the clamp.
- Zen Master
- Posts: 435
- Joined: Thu Jun 07, 2007 10:53 am
- Location: Cambridgeshire
Return to USB PC Data Loggers
Who is online
Users browsing this forum: No registered users and 0 guests