Post any questions you may have about our current range of USB data loggers
I am using two TC-08 data logger connected to the same pc with two different USB ports. Please can you explain me how to read all 16 temperatures at the same time with pico log software? Just one item is recognized by the software, now.
Settings > Recordings > Check in 'Use multible Converters' > OK
Settings > Input Channels > Add...
All the menus have a Help button that brings you direct to help on the topic.
I use two TC-08's and it worked 'right out of the box'.
/ Best regards
- Active User
- Posts: 7
- Joined: Sat Jan 01, 2005 9:39 am
- Location: Copenhagen, Denmark
Return to USB PC Data Loggers
Who is online
Users browsing this forum: No registered users and 1 guest