Post any questions you may have about our current range of oscilloscopes
Using picoscope6 how do use the signal generator? There is no amplitude control and the signal out of the EXT is distorted and low amplitude. What am I missing?
The output of the 3206 signal generator is a fixed amplitude (+/-0.5V) at an impedance of 600 ohms. If fed into a 50 ohm load, the voltage will drop to about +/- 40mV but it should not be distorted.
- Zen Master
- Posts: 435
- Joined: Thu Jun 07, 2007 10:53 am
- Location: Cambridgeshire
Return to USB PC Oscilloscopes
Who is online
Users browsing this forum: No registered users and 0 guests