Having problems ? let us know the details here
Can someone please explain how i change the sacale of the graph and other information in Evironmon when using the 10mV/A setting on the current transudcer?
i can get measurement perfectly when using the 1mV/A setting but on the 10mV/A all data is multiplied by 10, i want to use this setting in a lower current ranges so need to be more sensitive.
There is no way to alter the scaling in the current converter. It was originally designed for a current clamp that only had one setting. Later clamps came with a x10 setting but the converter and software know nothing about that! You will have to make a mental note to adjust all readings later, Sorry!
- Zen Master
- Posts: 435
- Joined: Thu Jun 07, 2007 10:53 am
- Location: Cambridgeshire
Return to Getting Started
Who is online
Users browsing this forum: No registered users and 0 guests