My name is Pedro Bertoleti and I work at LAT-EFEI (Itajubá-MG, Brazil).
I need some help with a PicoTech product. Here we have a PC Osciloscope ADC-212 and we have some problems is its data resolution. We developed a software in C programming language (based on the PicoTech Software Development Kit) and we are measuring with "trigger" (we set the trigger level, timebase, delay percent and interval between 2 consecutives measurements). The results of all voltage measurements are integer values, and for our application it isn´t a enough voltage resolution. We use the PicoScope software to compare the results and the problem stand still in PicoScope Software.
Are there possibilities to increase this voltage resolution to a bigger (float values) resolution?
We also verified that the maximum measurement points number is 2500. Is there a possibility to increase that points number in ADC-212?
We are buying a new PicoTech product (a PicoScope 3205). Do these problems I wrote (points number and voltage resolution) happen in PicoScope 3205?
If its possible, may you send me a complete documentation of ADC-212 and PicoScope 3205?