Post any questions you may have about our current range of oscilloscopes
just had delivery of a 3204 today, channel A works ok but there isnt a trace for channel B. Is there something in the software that I havn't set or is the unit faulty?
- Posts: 0
- Joined: Thu Jan 07, 2010 6:52 pm
Have you setup Channel B in the software? On software startup it defaults to having Channel A on with the rest of the channels off. You will need to select the proper voltage range you are going to be looking at. Or is the trace displayed but it does not change when you apply a signal to the scope?
Crag Technologies, Inchttp://www.pc-oscilloscopes.com
- Posts: 109
- Joined: Thu Sep 28, 2006 3:50 pm
my channel B shows me overvoltage when i applyed nothing on the input what should i do
is it broken?
- Posts: 0
- Joined: Mon Mar 15, 2010 10:05 am
If you have any suspected fault with the hardware it is best to contact us directly on helpdesk:-firstname.lastname@example.org
We can sort it out there.
- Zen Master
- Posts: 435
- Joined: Thu Jun 07, 2007 10:53 am
- Location: Cambridgeshire
Return to USB PC Oscilloscopes
Who is online
Users browsing this forum: No registered users and 1 guest