Input Range

Post any questions you may have about our current range of oscilloscopes
Steven

Input Range

One of the specifications of the PicoScope 2205 is:
±50 mV to ±20 V in 9 ranges
Those this mean that the minimum voltage that can be detected and displayed is 50mV?

And what does the 9 ranges mean?

PeterF
Posts: 435
Joined: Thu Jun 07, 2007 10:53 am
Location: Cambridgeshire

Re: Input Range

Hi,
The most sensitive range on the 2205 is ±50 mV. That means that since it is an 8 bit scope it has 256 steps of resolution over the full range from -50 mV to + 50 mV and can resolve in steps of 100 mv/256 = 0.39 mV. A sine wave that is 4 mV peak to peak will be a bit like "Steps" with only 9-10 individual voltage levels.
There are 9 such ranges available:- ±50 mV, ±100 mV, ±200 mV,±500 mV, ±1 V, ±2 V, ±5 V, ±10 V, ±20 V.
Each range will have a voltage step equal to the full scale deflection (±) divided by 256.
PeterF.

Picoscopian
Newbie
Posts: 0
Joined: Fri Nov 06, 2009 2:46 pm

Re: Input Range

I would highly appreciate if you could add ranges for asymmetric signals.

E.g.

-1 to 9V
0 to 10V
-1 to 19V
0 to 20V

Many thanks

PeterF