Forum for discussing PicoScope version 6 (non-automotive version)
If I send a 1 kHz sine wave on my 2205 from the wave generator to Chan A or B, I measure there a 2.5 kHz Signal.
Tried time measurement with our calibrator, but can't get right timing results?
- Posts: 0
- Joined: Fri Nov 06, 2009 2:46 pm
Yes, we have found a bug with the Beta software 6.3.26 for just the 220x series. The frequency output of the generator is x2.5 higher than set. Please go back to the standard release 6.2.20 available here:-http://www.picotech.com/software.html
- Zen Master
- Posts: 435
- Joined: Thu Jun 07, 2007 10:53 am
- Location: Cambridgeshire
Return to PicoScope 6 for Windows
Who is online
Users browsing this forum: email@example.com and 0 guests