I am new to Picoscope (just bought the 4224). Is there any information on exactly how the software's built-in functions computes AC RMS and True RMS? Is Picoscope's True RMS really AC+DC True RMS? My application is to determine puwer out of an audio amplifier when driven by a signal of pink noise. AC-DC True RMS is needed.
I want to add a formula and display trace for AC+DC True RMS Volts and then AC True RMS Volts without the DC offset and finally the DC offset by itself.
Then I need to do the same for the current measurements.
Then I can calculate power to the speakers with and without the DC component.
The application is to characterize non sinusoidal, irregular audio band signals going through a Class-D switching audio amplifier. The amps output is differential so I will tie up two probes that will have to be added together before the math functions of AC +DC True RMS are performed.
In Picoscope 6 there does not seem to be way to have the Math channels (or the measurement perform an integration of the data to allow me to set up the formula for calculating AC+DC True RMS. I would need to perfrom this for all data points in the sample and then calculate an average. Then I do this for the current and then I can mutiply to get the power result.
Also, when taking current reading by measuring voltage across a low value in-series resistor, how should I approach setting up the scope? Should I set the resistor/current measuring network as a custom probe and have the scaling function do the divide by the esistor value to get current then integrate each point to get AC+DC volts RMS?
Tnanks in advance for you help folks.