Post any questions you may have about our current range of oscilloscopes
I'm using a 2205 oscilloscope to analize analog signals in the range of 0-3V. I've notice that there is no way to waste half of dymamic range of the oscilloscope A/D converter. Infact if I use the +/-2V range I see the overange indication.
Is there any way to adjust the offset and maintain full dynamic range (i.e. subctracting a DC component)?
- Posts: 1
- Joined: Mon Jun 08, 2009 8:10 pm
The 2205 has nine input ranges, from Â±50 mV to Â±20 V. It is not possible to change these.
You can try using resolution enhancement to reduce the effect of quantisation in the waveform.
- Zen Master
- Posts: 558
- Joined: Fri Sep 19, 2008 10:17 am
Return to USB PC Oscilloscopes
Who is online
Users browsing this forum: No registered users and 0 guests