We are using our 3224 scopes for vibration spectrum monitoring on turbine engines. From calibrating our Data System we know that 66.8mv equals 1 IPS (Inches per second). We set up a math channel with the formula (channel)/66.8 to tell us in IPS what the given amplitude is at the specific frequency.
As an example: 20mv = .2994 IPS
What we are getting though is values in scientific notation.
Are we missing something in the setup? Is there a way to force the program to only use straight decimals?
Knowing our luck, it's probably something simple and stupid that we are missing.
Math Channel Setups
Math Channel Setups
 Attachments

 PicoScope Print.pdf
 PDF showing what we see when using the math channel markers
 (34.04 KiB) Downloaded 213 times
Hi
The ruler measurements take the same form as the yaxis labels. For small numbers, standard form is used (to maximise graph space) and it's not possible to change this.
If you use a custom probe to do the scaling instead of a maths channel, it would use unit prefixes rather than standard form. For example, your yaxis might be in mIPS or microIPS.
Robin
The ruler measurements take the same form as the yaxis labels. For small numbers, standard form is used (to maximise graph space) and it's not possible to change this.
If you use a custom probe to do the scaling instead of a maths channel, it would use unit prefixes rather than standard form. For example, your yaxis might be in mIPS or microIPS.
Robin