as part of a bigger VI I want to measure several times 10 triggered pulses with my PS5444B. I use the rapid block mode for this with the GetValuesBulk function.
For some reason this measurement takes several minutes instead of the expected 10's of seconds. I noticed when running the VI that it always stalls at the "isReady" call. What would cause the stall at "isReady"?
Attached is a simplyfied version of the relevant part of the VI. It is quite similar to the example VI, which works fine for the same settings.
The signal for testing the VI is a 4V, 10Hz square signal (see attached .png).
Thanks in advance
Post your LabVIEW discussions here
2 posts • Page 1 of 1
- (17.26 KiB) Downloaded 133 times
The problem was easily fixed by connecting the "Max ADC"- value properly to the "settings"-VI...