I have noticed strange behaviour of my LV code with picoscope drivers.
From time to time the measurement of pulses shows 4V level instead of 10V. I can simulate this with turning off the channel measurement (error 5070 at ps4000aGetValues.vi PICO_BUFFER_NOT_SET):
then turning it on again:
In order to fix that I need to change Range from 20V to 10V (or other), and then back again to 20V.
I can simulate this, but it also happens without any reason, randomly on random piscoscopes (I have 2 connected).
I am using: Windows 10, LabVIEW 2017, ps4000a.dll that comes with newest PicoScope 6 application.
Post your LabVIEW discussions here
1 post • Page 1 of 1