I have recently upgraded my PC to Vista, and have downloaded the latest Pico software (5.20.0).
The display does not register with the trigger position properly. It seems to be shifted to the right.
For instance, I have a timebase of 1ms/div and a trigger delay of -10%. What happens is that the trace starts at -142.8 us and the waveform which caused the trigger starts at 859.3 us. This adds up to 1 ms (within a bit of tolerance). The trace should start at -1 ms and the waveform should start at 0 ms. So I get a bit of blank trace at the left of the screen, and some of the post trigger on the right is lost.
I have attached a screen shot which shows the problem.
The scope is an ADC212-50 with a parallel to USB interface. It has been running OK for years on W 98 and XP.
Is this a PIco Vista problem?