Since installing 184.108.40.206 I have had a bit of trouble with both Picoscope and Picolog. My current problems:
Frequently when I run picoscope it can't find the hardware. When I unplug and plug the 3424 in again, XP wants to reinstall the software. I let it do that today. Picoscope could then find the 3424 and display a signal, although occasionally the first window it put up said it couldn't find the hardware, but when I closed this window, there was another window underneath with the oscilloscope running. When Picoscope is running the light is on on the 3424, but when I close Picoscope, the light goes out.
With Picolog I also have trouble connecting to the 3424, but I'm not sure it dated from the 220.127.116.11 installation. But this is not the most serious problem. Today when it is connected and I open a file to record data and set it up for data, before pressing the record button the signal indication is fluctuating indicating a signal is present, but the light on the 3424 is out. When I press the record button, the light on the 3424 comes on, but the signal indication stops fluctuating. The signal in this case is only a few millivolts of noise averaging around zero millivolts. After recording the data record is zero mVs throughout the record.
I suspect a hardware problem in the 3424 unless it is something the 18.104.22.168 software has done. Do I have a one year warranty on the 3424?