Thanks for your reply.
I have another doubt. I've been using the PicoScope 6 software in DEMO mode to find out how it works. From the figure 2 on the web page www . picotech . com / deep-memory . html , I can depict that selecting the 1 s/div timebase I would be able to get a sample rate of aprox 6 to 8Ms, but selecting this timebase in the DEMO software the properties dialog shows that the sample rate is only 244Ks/s. Is this difference due to the DEMO software? Will it be possible to obtain the 6 - 8 Ms/s in the 1s/div timebase with the real 5204?
I'd also like to know if it's possible for the 5204 to work as a signal generator and scope simultaneously, i.e. output an arbitrary waveform on CH1 and capture it on CH2 for example.
Thanks for your assistance.
DELPHI electronics and safety.