the specifications of picoscope 3424 are: 20 MS/s (1 channel)
I'm using 1 channel for my signal and 1 channel for my trigger. When I analyse my data I notice that there is a datapoint every 5 ns (setting 500 ns/div). Are all these points real datapoints, because one would expect a datapoint every 50 ns: 1 divided by 20 MS/s equals 50 x e-9 s?
Or thus the programme interpolates this extra datapoints?
I'm using the latest software r5_16_2.
By the way: I'm happy to see that the microvolt resolution problem when using save as txt is not there anymore in the new software!
Forum for discussing PicoScope (version 5)
2 posts • Page 1 of 1