Hi and thank you for your email.
The TA009 current clamp has an output of 10mV per amp if you were to choose the 60A setting or 100mA/V if you were to choose the 20A settings. So you have to do some scaling using a look up table.
For example for the 60A settings using the custom ranges option you would put something like
This shows three pairs each pair has the raw value (in mV) then the scaled value, also you can add units to it. Please see the attached file.
Then from the voltage range input you can select it as shown on the attached image.
- custom ranges2.JPG (21.46 KiB) Viewed 1975 times
- custom ranges.JPG (16.94 KiB) Viewed 1978 times