When I make a zero check on the two channels of my ADC-212/50, I see an offset.
This offset varies proportionally to Voltage range. Moreover, this offset doubles when I use Trigger on channel A.
Have you an explanation to that?
Thank in advance, Blondy.
Forum for discussing PicoScope (version 5)
2 posts • Page 1 of 1