Post any questions you may have about our current range of oscilloscopes
- Posts: 0
- Joined: Wed Sep 20, 2017 8:20 pm
How can I see the complete measurement in 1 window, instead of 31 buffers for example. I have this issue that between buffers data is lost. Example, the signal at the end of the buffer is -1V, the next buffer it starts with -3V.
I want to see the complete measurement. Any tips?
- PICO STAFF
- Posts: 527
- Joined: Mon Aug 11, 2014 11:14 am
What is the sampling frequency that you're using now, and what is the slowest sampling frequency that you can tolerate for the captures?
- Site Admin
- Posts: 187
- Joined: Thu May 25, 2006 8:02 am
Rather than collect 32 short buffers (eg of 1ms/div) try collecting one longer one (eg 20ms/div). You can then zoom in and see all the detail in one continuous block.