I used ADC11-10 and Picolog R.16 version, real-time continuous mode with sample rate=100ms, scan time=20000us.
I met a quite strange problem, when I scaling the load cell voltage to physical units(force in Newton) using scaling function in Picolog while recording that load cell voltage at the same time. I.e., I set two measurements using the same input channel, one is for original voltage output, the other is for scaling the voltage to force. I found that the scaled force data doesn't have high linear correlation with my recorded voltage data(R^2=0.77), it should be 1 since I just used the linear scaling such as force=voltage*10.
I further check the screen reading for scalled force value and original voltage value, it showed that they are not changed in the same time!! This makes me feel very frustrated. does anyone meet the similar problem as me? Is it the bugs in the Picolog? Why this happened?
I am eager to hear some solutions or suggestions! Thanks.
Forum for discussing PicoLog
2 posts • Page 1 of 1
- Advanced User
- Posts: 656
- Joined: Thu Jul 07, 2005 12:41 pm
- Location: St Neots, Cambridgeshire