There is no facility to take the settings from the PicoScope 6 software and directly apply them via the driver API functions.
I would suggest looking at the settings that you have in the PicoScope 6 software, i.e.
- channel settings (range, DC coupling and offset)
- number of samples
- trigger settings
- sampling interval (via the Properties pane)
You can then map these to the appropriate settings in the LabVIEW example, referring to the Programmer's Guide
If you require further assistance with this, please send the PicoScope settings file (.pssettings) or a data file showing the actual signal and we can advise further.