I 've just bought the 3206, and my problem is :
When i shortcut probe with ground, my chanel A have 1,6v and my chanel B have 0v. if I reverse the probes, I have the same thing on channel A.
I can't remove offset with 5.13.9 picoscope software in settings menu. No choice chanel appeared.
What can i do ?