Post any questions you may have about our current range of oscilloscopes
Our Pico 3206 jus work out. Channel A give no response (channel B still allright). It's shows on picosiloscope, that channel A have yellow ! mark. Even affter the voltage limit set to 20 Volt.
You will need to return the unit to us for repair.
Please email your details to firstname.lastname@example.org
Michael - Tech Support
Web Support Forum
- Zen Master
- Posts: 656
- Joined: Thu Jul 07, 2005 12:41 pm
- Location: St Neots, Cambridgeshire
Return to USB PC Oscilloscopes
Who is online
Users browsing this forum: No registered users and 0 guests