AndrewA wrote:Hello Maziar,
The 3406D has enough analog bandwidth to for the decoder work correctly. Even our 6404D and 6407 scopes will be limited by size and number of packets when decoding. These models could used to look at the eye diagrams, but one of our 9000 series units would be the best for eye diagrams.
Currently for the 100base-tx decoding to work correctly, you need to set 'between rulers' or limit the number of samples, to be no more than 20MS. This is a bug in PicoScope 6.11.10.
Are you want to just decode the packets, or look at the signal quantity? - eye diagrams?
If you want to decode the packets, I would use a PC/laptop using the NIC and LAN tap and run TCP/IP decoding software like wire-shark, which is free. https://www.wireshark.org
LAN tap- https://greatscottgadgets.com/throwingstar/ Or make something with the same connections.
Users browsing this forum: No registered users and 0 guests