I’m searching for a current- Logger for logging 0-15 Amps, and now I have got a cm3 for testing purpose.
But the logger-response time is awful - after reducing current from 10 to zero ampere, 30 seconds later it still shows about 2 ampere!
I did not find any information about this in the specification, but on your website you are showing an image with much faster response. Is there a way to adjust the averaging time? (Otherwise it does not meet my expectations)
BTW: one of the three clamps doesn’t close completely, this causes a difference of 10% between the three chanels.