I think that I am doing something wrong. I am running PicoScope 5.09.4 on a Win 2K laptop with an ADC212-100.
When running both channels and using the cursor measurments, I only seem to get results on the "A" channel. Is there a way to make the cursors measure both A & B at the same time?
P.S. I LOVE MY PICO!!! I love the functionality of having data already in the computer where I am going to process it anyways in 3rd party apps. I use Word for reports, Excel for reports and charts, MathCAD/Matlab for curve fitting. I embarace co-workers who spend $3,000 U.S. dollars on a "cheep" Tektronics scope. They bought a fine instument, but my Pico reports data intantly ready for procesing in external applications. I do this every day as an electronics technician and a Elec.Eng. student in labs. Other scopes make you jump through hoops like a monkey just to report on raw ASCII data. I can't wait for your future upgraded products. This is one happy camper.