Forum for discussing PicoScope version 6 (non-automotive version)
I've been using 5244B for performance measurements of an amplifier used for position controller. Looking at the noise levels, in Spectral Mode, there are situations where the baseline of the spectral plot (x-axis) is chopped off. There does not seem to be a way to set the vertical axis range from min to max in order to see the baseline. Possibly this is a resolution problem, maybe the signal falls below the noise floor of instrument. But the behaviour seems inconsistent. See attached print that shows what I mean. Is there a way to control the vertical range displayed in spectral mode?
- PicoScope 5244B spectral display anomaly 6-27-14.pdf
- screen shot showing bottom chopped off
- (94.59 KiB) Downloaded 56 times
- Posts: 0
- Joined: Sat Jun 21, 2014 12:19 am
As its just off the screen try dragging up the left hand axis with the mouse to reveal more. Another option is to click on the x1 icon at the bottom left, from this you can set a zoom of say 0.9 to reveal more.
If that does not work, let us know and we will find another solution.
- Site Admin
- Posts: 173
- Joined: Thu May 25, 2006 8:02 am
Return to PicoScope 6 for Windows
Who is online
Users browsing this forum: No registered users and 0 guests