i have some problems with a Tektronix P6015A High-Voltage Probe and a PicoScope 5244B. When i connect the Probe to the Oscilloscope it works pretty fine. But when there is a voltage precisely in the range of 2kV pk-pk the Oscilloscope drops out with red light and loosing usb connection. Switch off and on solves this problem till the next 2kV peak.
The P6015A is a passive 75MHz probe with 1:1000 so with 2kV pk-pk it puts out about 2V so there should be no problem for the Oscilloscope.
Any ideas why this happens?