Post any questions you may have about our current range of oscilloscopes
I just purchased a Picoscope 5242A and noticed that the signal generator is very imprecise for low frequencies below 5Hz and not working at all below 200mHz in Picoscope 220.127.116.11!
I also tried to use the C code provided with the SDK but can't generate slow signals(Should work as the Specification says from DC to 20 MHz).
What am I doing wrong?
- Posts: 0
- Joined: Mon Nov 25, 2013 10:27 am
Could you save your settings in PicoScope6 and then email the ps data file to email@example.com
and we can investigate further for you.
- PICO STAFF
- Posts: 7
- Joined: Tue Jun 04, 2013 12:07 pm
After investigation, it appears that there is a bug which will be submitted to our Developent Team.
The minimum resolution should be 50mHz.
- Site Admin
- Posts: 2171
- Joined: Tue May 31, 2011 3:43 pm
- Location: St. Neots, Cambridgeshire
Return to USB PC Oscilloscopes
Who is online
Users browsing this forum: No registered users and 0 guests