I am trying to use the ADC200/20 with PicoScope 5.11.6 just as I would use a conventional oscilloscope. I need to set the PicoScope for external horizontal drive with a vertical sensitivity of 1V/CM. How can this be done? My application is to use the Dual Sweep method of FM receiver alignment using a Sound Technology ST1000A FM alignment generator. I need a 60Hz sinewave to control the horizontal drive of the Pico so I can create an envelope wave form on the display. Any suggestions would be marvelous. Happy Holidays to all.
Vacuum Tube Valley NYC Technical Affiliate