Having problems ? let us know the details here
I purchased 2ea. usb tc-08 temperature loggers and I am currently reading a total of 13 thermocouples. I would like to connect 2ea. vacuum transducers to one of these units. How can this be accomplished and which model of transducer
do I need to purchase? The units are currently being used in a composite oven and we need to have the vacuum recorded in
inches of hg.
- Posts: 0
- Joined: Sat Jun 01, 2013 1:49 am
The single channel terminal board allows the connection of 4 - 20mA sensors or voltages from 0 - 5V to be input into the TC-08:http://www.picotech.com/document/pdf/DO167-USB-TC08-Terminal-Board-Users-Guide.en-1.pdf
Unfortunately, I'm not able to advise on transducers - you could try an electronics specialist where you are located.
You would have to set up a scaling file in the PicoLog Recorder software.
I hope this helps.
- Site Admin
- Posts: 2058
- Joined: Tue May 31, 2011 3:43 pm
- Location: St. Neots, Cambridgeshire
Return to Getting Started
Who is online
Users browsing this forum: No registered users and 0 guests