Having problems ? let us know the details here
I'm looking to buy a Picolog 1216.
How can I do a Cold junction compensation on these data loggers?
- Posts: 0
- Joined: Mon Mar 11, 2013 6:18 pm
The Picolog 1216 is not designed for use with thermocouples, it needs external circuitry to make it work, so a TC-08 would be a better option.
That said the following document has a section on connecting a thermocouple to another of our products the ADC24 http://www.picotech.com/document/pdf/adc2024_terminal_board_user_guide-5.pdf
although I would suggest contacting firstname.lastname@example.org
for further information
- Site Admin
- Posts: 2338
- Joined: Fri Jun 10, 2011 8:15 am
- Location: St. Neots
Return to Getting Started
Who is online
Users browsing this forum: No registered users and 0 guests