Post any questions you may have about our current range of USB data loggers
What is the maximum common mode voltage when using the ADC24 in differential mode?
I need to monitor individual cell voltages in a 5 cell series string but the 4th and 5th cell readings are incorrect - what is wrong?
- Posts: 0
- Joined: Thu Sep 20, 2012 10:46 am
Channel to common ±1.9 V (±39 mV to ±1250 mV ranges)
Channel to common ±3.0 V (±2500 mV range)
Common to earth ground ±30 V
- Site Admin
- Posts: 2355
- Joined: Fri Jun 10, 2011 8:15 am
- Location: St. Neots
Return to USB PC Data Loggers
Who is online
Users browsing this forum: No registered users and 0 guests