I have a PicoScope 3205B which i'm trying to use with PicoLog. But when I'm choosing PicoScope 3000-series in the setup, no devices appears. When a press the Refresh-button the programs searches a second or two, but dosn't find anything. I'm currently not running PicoScope and the oscilloscopes appears in the device manger. I have no problem running PicoScope 6 with my device.