Forum for discussing PicoLog
- Posts: 0
- Joined: Thu Aug 16, 2012 8:22 am
I am using the picolog CM3. Have tried connecting the the clamps to several 3 phase supplies but the reading dont higher than 0.1 A. Can anyone help me please?
- Site Admin
- Posts: 3583
- Joined: Fri Jun 10, 2011 8:15 am
- Location: St. Neots
Ensure that you are clamping around the individual live wire, not the overall cable.
Technical Support Manager