Post any questions you may have about our current range of oscilloscopes
Using PicoScope 22.214.171.124 with 2205MSO:
Channel A monitoring a fixed 5.1V signal shows correct voltage on both scope and measurement when the timebase is selected at 100ms/div or faster.
At timebase of 200ms/div or slower both the scope and the measurement show 5.77 volts with the same 5.1V input signal.
- Posts: 0
- Joined: Sat May 12, 2012 6:11 am
Hi I have recreated your problem, this appears to happen when in block mode, I have reported it as a bug.
The beta version of the software does not have this problem.http://labs.picotech.com/#psw6
- Zen Master
- Posts: 1705
- Joined: Fri Dec 01, 2006 10:03 am
- Location: St Neots
Return to USB PC Oscilloscopes
Who is online
Users browsing this forum: No registered users and 0 guests