Post any questions you may have about our current range of oscilloscopes
Hi, I recently purchased a Picoscope 2207.
There seems to be something wrong with the measurements.
When measuring a standard USB bus, the output on the screen looks like a sin wave (goes negative as well).
Voltage is also close to 12volt.
I don't understand this.
I am using the PP821 probes.
- Posts: 0
- Joined: Thu Apr 05, 2012 12:32 pm
You will see a sine wave by default if the PicoScope software hasn't detected the 2207 and is running in demo mode.
To see if this is the case go to Help->About and check the Model: SerialNumber: and DriverVersion: information.
- Site Admin
- Posts: 2400
- Joined: Fri Jun 10, 2011 8:15 am
- Location: St. Neots
Return to USB PC Oscilloscopes
Who is online
Users browsing this forum: No registered users and 0 guests