Post any questions you may have about our current range of oscilloscopes
I'm doing rise time measurements using an external generator. Square wave is 2kHz, 10Vpp.
80/20% measurement tells me 400ns but 90/10% says 153us (!)
Adding cursors to the graph indicates 680ns for the 90/10%.
Why is the 90/10% measurement so far out?
PicoScope® 6 - PC Oscilloscope software version: 22.214.171.124
Model: PicoScope 4262
Driver Version: 126.96.36.199
Firmware Version: 188.8.131.52 / 184.108.40.206
- Posts: 0
- Joined: Sat Mar 03, 2012 12:20 pm
- Location: Norway
We are currently looking at the Picoscope 6 measurements as there may be a few inaccuracies with these.
- Site Admin
- Posts: 2585
- Joined: Fri Jun 10, 2011 8:15 am
- Location: St. Neots
I am seeing the same issue, is there any updates on this issue ?
- Posts: 0
- Joined: Wed May 30, 2012 6:39 pm
HI I have investigated this myself and there is a problem with the 10- 90% calculations for rise and fall time. The software engineers are aware of it and we will be looking at fixing this.
- Zen Master
- Posts: 1705
- Joined: Fri Dec 01, 2006 10:03 am
- Location: St Neots
Return to USB PC Oscilloscopes
Who is online
Users browsing this forum: No registered users and 0 guests