I am seeing a strange glitch with my new 2205 MSO in Picoscope in very specific circumstances. If I view an analog waveform with either the A or B channel, with the pre-trigger set to 10% or 30%, and I adjust the collection time to any rate that results in 10,005 samples (typically between 5 and 20 us/div, depending on which probes are turned on), then the last 80% of the wave (from 20% all the way to the right edge of the screen) appears as nothing but a flat line. This happens regardless of whether I use Auto, Repeat or Single trigger modes. In Rapid mode, the wave goes flat at 10% or 20%, regardless of what the pre-trigger is set to, for any collection time that results in 10,005 samples or more. Any idea what might be going on here?