I have a PicoScope 3204, win XP sp2
I find that even with the probe shorted my scope see a noise of about 8,10 mV pp
With this amount of noise, the scope is useless Am I doing something wrong?
Thanks for your help,
You can easily see 5mV ptp noise as it is only an 8 bit scope with a least significant bit of 790uV. If you wish to see less noise you must reduce the bandwidth being measured, either by enabling a low pass filter or using some "Resolution Enhancement", both available in the channel options menu of PicoScope6.
Users browsing this forum: No registered users and 0 guests